# What does Analog Multimeter Sensitivity mean?

What does Anlaog Multimeter Sensitivity mean?

Please provide an example of how Analog Multimeter Sensitivity is specified?

How are analog meters useful for gaining an intuitive understanding of instrument sensitivity?

Thanks!
LVL 1
###### Who is Participating?
I wear a lot of hats...

"The solutions and answers provided on Experts Exchange have been extremely helpful to me over the last few years. I wear a lot of hats - Developer, Database Administrator, Help Desk, etc., so I know a lot of things but not a lot about one thing. Experts Exchange gives me answers from people who do know a lot about one thing, in a easy to use platform." -Todd S.

Commented:
Analog multimeter sensitivity - explained

One of the specifications for an analogue multimeter or VOA meter is its sensitivity. This comes about because the meter must draw a certain amount of current from the circuit it is measuring in order for the meter to deflect. Accordingly the meter appears as another resistor placed between the points being measured. The way this is specified is in terms of a certain number of Ohms (or more usually kOhms) per volt. The figure enables the effective resistance to be calculated for any given range.

Thus if a VOA multimeter had a sensitivity of 20 kOhms per volt, then on the range having a full scale deflection of 10 volts, it would appear as a resistance of 10 x 20 kohms, i.e. 200 kohms.

When making measurements the resistance of the meter should be at the very least ten times the resistance of the circuit being measured. As a rough guide, this can be taken to be the highest resistor value near where the meter is connected.

Normally the sensitivity of an analogue meter is much less on AC than DC. A meter with a DC sensitivity of 20 kohms per volt on DC might only have a sensitivity of 1 kohm per volt on AC.

Experts Exchange Solution brought to you by